Are you looking for the best brand new battery at the most preferred price? Here is the right place! We provide the highest quality boolclass for the lowest price with the highest level of service, all in a secure and convenient platform. Our product range include replacement Notebook battery, Cordless Drill batteries, Digtal batteries and Biomedical batteries for most major brands, including Dell, Apple, Compaq , HP, IBM , Lenovo, Fujitsu, Gateway, Sony, Toshiba, Nihon Kohden and GE.
Just order here , Buy With Confidence:
High Quality: We believe that you will get the new highest quality batteries and adaptes at the lowest prices. We continuously test many brands of Biomedical batteries to ensure that we get the best quality batteries for you.
Low Price: As your see, most batteries ans adapters have an up to 30% Discount. High Quality, Low Price, 100% Guarantee, Fast Shipping, No Sales Tax. Just order what you want.
Best Service: You want it, we got it. We have a dedicated team of customer service executives who will tell you what you want to know and give your the best shopping experience.
Fast Shipping: We shipping Philips Suresign VS3 ECG EKG Battery the same day, and you can choose two shipping ways. The faster way will only take 2-5 days Worldwide. Please leave a recipient's phone number for Secure delivery at first.
Thank you, Happy shopping!
Good Tips For Getting Maximum Shelf Life For Your Suresign VS3 Battery:
Your new battery comes in a discharged condition and must be charged before use (refer to your Medical Equipment for charging instructions). Upon initial use (or after a prolonged storage period) the battery may require two to three charge/discharge cycles before achieving maximum capacity.
When charging the battery for the first time your Medical Equipment may indicate that charging is complete after just 10 or 15 minutes. This is a normal phenomenon with rechargeable batteries. Simply remove the Philips Suresign VS3 battery from the Medical Equipment and repeat the charging procedure.
It is important to condition (fully discharge and then fully charge) battery every two to three weeks. Failure to do so may significantly shorten the battery's life (this does not apply to Li-Ion batteries, which do not require conditioning). To discharge, simply run your device under the battery's power until it shuts down or until you get a low battery warning. Then recharge the battery as instructed in your user's manual.
If the batteries will not be in use for a month or longer, it is recommended that it be removed from the device and stored in a cool, dry, clean place.
It is normal for replacement battery to become warm during charging and discharging.
A charged battery replacement will eventually lose its charge if unused. It may therefore be necessary to recharge the battery after a storage period.
Note:Most Medical Equipment of Nihon Kohden and GE should set up again after replaced the battery. The battery data should cleared or reset, otherwise your machine will warn error. That not mean the battery is no good. Please set up your machine and go on use the batteries.